Model Number: SSI 521 HSH
Standard Number: IS 7098 (Pt.1, 2 & 3), IS: 10810 (Pt. 30)
Laboratory Hot Set Test Apparatus is designed to determine the elongation & permanent set of insulation of armoured and unarmoured single, twin, three , four and multi-core cross linked Polyethylene (XLPE) insulated & PVC sheathed cables for electric supply and control purposes under specified conditions of load temperature and air flow.
Laboratory Hot Set Test Apparatus is designed to determine the elongation & permanent set of insulation of armored and unarmored single, twin, three , four and multi-core cross linked Polyethylene (XLPE) insulated & PVC sheathed cables for electric supply and control purposes under specified conditions of load temperature and air flow, as specified in IS: 7098.
The instrument conforms IS 7098 (Pt.1, 2 & 3), IS: 10810 (Pt.30)
A double walled cabinet with inside chamber of stainless steel
Inner dimensions of the chamber are 30 cm x 50 cm 20 cm (W x H x D).
Full view glass Door to observe the specimen under test.
Temp. Range 250 ± 2oC. at sensor point.
The oven is designed to give 8 to 20 air changes per hour.
An Digital air flow meter is fitted to monitor the airflow rate. Range 0- 10 LPM, LC-0.1 LPM
Grips are provided at the top of the oven to hang specimen/s.
Suitable weights up to 1 kg are provided with provision to attach the weights to one end of the test pieces.
Two ultra sonic sensors combined with laser pointers are given on the front door.
Laser pointers are attached to facilitate the operator to take exact reading of the sample elongation.
A blade is provided in the oven to cut the sample to remove the tensile force without opening the door.
The sample is hanged using the grips inside the Hot Set Oven along with the weights for 15 mins. The sample starts to elongate due to heat. The elongation of the sample is measured using vernier calipers. The sample is then cut to evaluate the permanent set.
The temperature and timer are controlled by a PLC. After the time period is over, the HMI prompts the operator for elongation measurement. The operator points at the upper and lower marks placed on the sample, and the ultrasonic sensors will calculate the distance traveled and pass the information to the PLC. The PLC deduct the original length of the sample to calculate the final elongation. To cut the sample, the operator can press the button on the HMI to operate the blade to remove the tensile force without opening the door.
The test requirement is 8 to 20 air changes per hour, measured on a Flow Meter. According to IS standard, if the Oven does not have a regulated air supply, the oven is then termed as a Hot Air Oven and not a Hot Air Oven (as offered by some manufactures) and cannot be used for this kind of test.
Also if normal oven is used for the test, the operator will have to open the front door to hang the samples, resulting in loss of temperature. As the test is only 15 mins, it will not be possible to regain the temperature. In our Hot Set oven, we use an especially designed Top Hanger which can be hanged from the top of the Hot Set Oven, without opening the front door. To achieve minimum loss of temperature, we provide two hanger assemblies. One is used as dummy that is used for achieving the desired temperature. In the meanwhile operator can assemble the sample, weights etc on the other hanger, which can be then put in the oven, just be replacing the dummy into the oven. This way the loss in temperature is not more than few degrees.